Your changes have been successfully saved
Medical Devices Sector
قطاع الأجهزة الطبية
المركز الوطني لبلاغات الأجهزة والمنتجات الطبية
mdprc 042 12 17 000
Lullaby Resus plus
Medical Device Identifier:
All Lullaby RESUS Plus Systems -Part number: 2070100-001 (Lullaby Resus Plus - NIST) & 2070100-002 (Lullaby Resus Plus- DISS)
Reason of Field Safety Corrective Action:
GE Healthcare has recently become aware of a potential issue related to inaccurate manometer reading of Lullaby Resus Plus devices when used for T-piece resuscitation. During the neonatal resuscitation process using T-piece on Resus Plus devices, the Manometer shows higher PEEP reading than actual maintained at the end of T-piece. Less than expected PEEP may lead to insufficient ventilation. There have been no injuries reported as a result of this issue.
GE Healthcare will correct all affected products at no cost to you. A GE Healthcare representative will contact you to arrange for the correction.
Source Ref. Number:
SFDA urges all hospitals that have devices subjected to recall, to contact the company.
Copyright © 2008 Saudi Food and Drug Authority. All rights reserved.