Your changes have been successfully saved
Medical Devices Sector
قطاع الأجهزة الطبية
المركز الوطني لبلاغات الأجهزة والمنتجات الطبية
mdprc 103 06 18 000
Alinity s System
Medical Device Identifier:
List Number (LN): 06P16‐01
Reason of Field Safety Corrective Action:
Abbott has developed Alinity s System software V2.0 (LN 04U76‐07) to address known issues and enhance the overall Alinity s System (LN 06P16‐01).
Your Abbott representative will be scheduling mandatory upgrades of your Alinity s System to install Alinity s System software V2.0 in the upcoming weeks. In the interim refer to Appendix A for necessary actions required until software V2.0 can be installed.
In addition to correcting these software issues, the software update includes a number of changes intended to improve performance and reliability.
Follow your laboratory procedures and please retain this letter for your laboratory records.
Medical supplies & Services Co.Ltd Mediserv
Source Ref. Number:
SFDA urges all hospitals that have devices subjected to recall, to contact the company.
Copyright © 2008 Saudi Food and Drug Authority. All rights reserved.