Your changes have been successfully saved
Medical Devices Sector
قطاع الأجهزة الطبية
المركز الوطني لبلاغات الأجهزة والمنتجات الطبية
mdprc 115 12 17 000
Hemoglobin A1c (HbA1c) Reagent
IVD Test Reagent/Kits, Hematology, Calibration, Hemoglobin
Medical Device Identifier:
List Number (LN): 4P52‐21
Multiple Lot numbers of the affected device are provided in the attached FSN.
Reason of Field Safety Corrective Action:
The purpose of this Product Correction letter is to inform you Fetal Hemoglobin (HbF)
interference occurs at a level lower than what is stated in the HbA1c reagent package insert.
• Please review this letter with your Medical Director.
• Take the information above into consideration when using the HbA1c reagent 4P52‐21.
• Complete and return the Customer Reply Form.
• Please retain this letter for your laboratory records.
Medical supplies & Services Co.Ltd Mediserv
Source Ref. Number:
SFDA urges all hospitals that have devices subjected to recall, to contact the company.
Copyright © 2008 Saudi Food and Drug Authority. All rights reserved.