Your changes have been successfully saved
Medical Devices Sector
قطاع الأجهزة الطبية
المركز الوطني لبلاغات الأجهزة والمنتجات الطبية
mdprc 186 09 12 000
Abbott Laboratories Inc
Reagent, ARCHITECT HAVAb-IgG
Medical Device Identifier:
List Numbers: 6C29-20 and 6C29-25 Lot Numbers: See Attachment
Reason of Field Safety Corrective Action:
The ARCHITECT HAVAb-IgG assay shows a shift in the serum and plasma results to higher S/CO values over time, potentially leading to reduced specificity and increasedfalse reactive results.
Refer to Attachment . Determine if you have any of the reagent lots listed in your inventory. Revise the expiration date for all reagent lots identified in your inventory per instructions of Attachment. After applying new expiration dates to reagent kits as listed in Attachment , discontinue use of those kits which are now outdated, and discard the material according to your laboratory procedures. Contact Abbott to order replacement material.
Medical supplies & Services Co.Ltd Mediserv, Riyadh, (01) 4780555
Source Ref. Number:
SFDA urges all hospitals that have devices subjected to recall, to contact the company.
Abbott Laboratories Inc.pdf
Copyright © 2008 Saudi Food and Drug Authority. All rights reserved.