Your changes have been successfully saved
Medical Devices Sector
قطاع الأجهزة الطبية
المركز الوطني لبلاغات الأجهزة والمنتجات الطبية
mdprc 012 01 11 000
Varian Medical Systems
Radiotherapy, 4D Integrated Treatment Console
Radiotherapy, planning, simulators: computer treatment planning
Medical Device Identifier:
4DITC all versions up to and including 8.8.17. For affected devices, See attached list
Reason of Field Safety Corrective Action:
This advisory notice is to provide you with updated information on multi-isocenter plan management and acquisition of couch parameters for multi-isocenter plans at the 4DITC.
Recommended User Action 1- Varian recommends that users use caution when acquiring actual couch positions for any plan.2- Varian recommends that each plan be reviewed for quality assurance purposes in accordance to set departmental or organizational policies. 3- Multiple plans within the same course would be the preferred method to accomplish this goal. 4- Varian recommends that the digital graticule be turned on when evaluating any image 5- If you are unsure how to proceed, please contact the Varian HelpDesk for advice. Worldwide e-mail addresses are listed on the attached file as well as internet support. Varian Actions: 1- Varian is notifying all affected customers with this document. 2-Varian has also chosen to update the procedure described in CTB-TR-735 to inform users of the behavior acquiring couch parameters for multi-isocenter plans. 3- The ability to handle multiple isocenter plans will be introduced with 4DITC version 10.
Varian Medical Systems, Riyadh (01) 2772126
Source Ref. Number:
SFDA urges all hospitals that have devices subjected to recall, to contact the company
Copyright © 2008 Saudi Food and Drug Authority. All rights reserved.