Your changes have been successfully saved
Medical Devices Sector
قطاع الأجهزة الطبية
المركز الوطني لبلاغات الأجهزة والمنتجات الطبية
mdprc 047 01 15 000
InterStim, InterStim II, Itrel3 and Enterra
Medical Device Identifier:
InterStim (Model 3023), InterStim II (Model 3058), Itrel 3 (Model 7425), and Enterra (Model 3116)
Reason of Field Safety Corrective Action:
This letter is to notify you of updates to labeling related to the cycling feature in some Neurostimulation devices used for sacral nerve stimulation, gastric electrical stimulation and spinal cord stimulation.
References to the use of cycling for improved battery longevity, when compared to continuous mode, are not accurate for every combination of programmable parameters. If cycling is being used solely to improve device battery longevity, the information in attachment should be considered.
Gulf Medical Co.
Source Ref. Number:
SFDA urges all hospitals that have devices subjected to recall, to contact the company.
Copyright © 2008 Saudi Food and Drug Authority. All rights reserved.