Your changes have been successfully saved
Medical Devices Sector
قطاع الأجهزة الطبية
المركز الوطني لبلاغات الأجهزة والمنتجات الطبية
mdprc 036 01 19 000
Varian Medical Systems
ARIA Oncology Information System for Radiation Oncology
Medical Device Identifier:
Affected Version(s) / Lot(s): 15.1, 15.5, 15.6
Reason of Field Safety Corrective Action:
Varian Medical Systems has identified a problem in ARIA Oncology Information System (OIS) for Radiation Oncology such that a user may inadvertently chart medical record information in an unintended patient’s chart.
-Varian recommends that Users close each patient prior to opening a new one. This will prevent this issue from occurring.
-Varian Medical Systems is investigating this issue. The timeline of a technical solution is currently unknown. Varian will contact users as a fix is available.
Varian Medical Systems Arabia Commercial Limited
Source Ref. Number:
SFDA urges all hospitals that have devices subjected to recall, to contact the company.
Varian Medical Systems.pdf
Copyright © 2008 Saudi Food and Drug Authority. All rights reserved.