Your changes have been successfully saved
Medical Devices Sector
قطاع الأجهزة الطبية
المركز الوطني لبلاغات الأجهزة والمنتجات الطبية
mdprc 081 06 15 000
Screwdriver blade Cross-Lock Mid 1.7, dental-end
Medical Device Identifier:
Lot (16688) Item (503004417)
Reason of Field Safety Corrective Action:
Internal investigation found that the Screwdriver blade (2.3 mm) was mislabeled. Following inspection of the product, it was determined that they were erroneously marked as Screwdriver blade Cross-Lock Mid 1.7, dental-end; Lot (16688), Item (503004417).
Assist your Zimmer sales representative with the quarantine of any affected product. Your Zimmer sales representative will remove the recalled product from your facility.
Ebrahim M. Al-Mana & Bros. Co. Ltd.
Source Ref. Number:
SFDA urges all hospitals that have devices subjected to recall, to contact the company.
Copyright © 2008 Saudi Food and Drug Authority. All rights reserved.