Your changes have been successfully saved
Medical Devices Sector
قطاع الأجهزة الطبية
المركز الوطني لبلاغات الأجهزة والمنتجات الطبية
mdprc 109 06 13 000
Therapeutic Radiation System, Digital Accelerator
Precise Treatment System
Medical Device Identifier:
All Digital Accelerators with Electrons and Beam Modulator, MLCi/MLCi2, Agility or Asymmetric Heads
Reason of Field Safety Corrective Action:
Some Digital Accelerators have increased electron applicator auto-tracking settings in the field, to values greater than the factory-set defaults. Increasing auto-tracking values from factory-set defaults goes against Elekta recommendations and can result in the system becoming non-compliant with IEC standards, in particular with the safety standard IEC 60601-2-1
Please adhere to the instructions and advice given in manuals and system dialogues, which explicitly state that a change to auto-tracking values could compromise compliance with safety standards and can cause increased unwanted radiation dose to the patient ( More details in attached file ).
Analysis Device for Medical Service and Scientific, Dammam, (03) 8174422
Source Ref. Number:
SFDA urges all hospitals that have devices subjected to recall, to contact the company
Copyright © 2008 Saudi Food and Drug Authority. All rights reserved.