Your changes have been successfully saved
Medical Devices Sector
قطاع الأجهزة الطبية
المركز الوطني لبلاغات الأجهزة والمنتجات الطبية
mdprc 061 11 17 000
3M Health Care Ltd
Bair Hugger Temperature Monitoring Sensors
Monitors, Physiologic, Temperature
Medical Device Identifier:
Model No. 36000; Lot Nos.: 17230, 17233, 17234, 17241 Formerly known as SpotOn Temperature Monitoring Sensors
Reason of Field Safety Corrective Action:
Programming Translation Error May Lead to Temperature Readout Lower than Actual Patient Temperature
Identify, isolate, and discard any affected product in your inventory. If you have affected product, verify that you received the October 17, 2017, Urgent Medical Device Recall letter and/or the October 19, 2017, Field Safety Notice letter and reply form from 3M. Regardless of whether you have affected product, complete the reply form and return it to 3M using the instructions on the form. Upon receipt of the form, 3M will provide your facility with replacement product, if applicable. Notify all relevant personnel at your facility of the information in the letter(s), and forward a copy of the letter(s) to any facility to which you have further distributed affected product.
Source Ref. Number:
SFDA urges all hospitals that have devices subjected to recall, to contact the company.
3M Health Care.pdf
Copyright © 2008 Saudi Food and Drug Authority. All rights reserved.